A Manual for Above-Knee Amputees
The knee-disarticulation prosthesis is very similar to the above-knee prosthesis, except for the lower part of the socket and the knee mechanism. Before the introduction of the present day polycentric knee units, sockets for the prosthesis were usually made of leather and metal hinges were used to attach the socket to the shin. This type of prosthesis is still preferred by a few preferred even though it is bulky and control of the leg during the swing phase is difficult. However, most prefer one of the polycentric designs where the knee mechanism can be installed within the shin due to its special design.
The major objection to the polycentric units is that the knee protrudes slightly beyond the front of the shank when the amputee is sitting or kneeling.
Leather sockets are held on by a lacing. Plastic sockets usually have a foam liner in the lower part for the bulbous end of the stump to slip by so as to keep the socket in place.
All of the instructions given about use of the above-knee apply equally to the Knee-Disarticulation prosthesis.
Go to next section
Go to previous section
Go to Table of Contents
Copyright 1996 - Alvin L. Muilenburg and A. Bennett Wilson, jr.
The electronic version of this manual is presented by
Dankmeyer, Inc., with the permission of the authors. This manual may not be redistributed in any form. In order to obtain printed copies of the manual for personal use or distribution, please go to the ordering information page.